Tuners for noise parameter measurement (Maury Microwave)
By combining a Maury Microwave Tuner and a PNAX network analyzer, it is possible to set up highly efficient measuring stations for S parameter, noise figure, noise parameter and intermodulation measurements:
The noise figure of an amplifier is typically measured by means of a noise source in combination with a spectrum analyzer. Normally, the established Y factor method will be used, where a noise source at the amplifier input is switched into two states: first into the OFF state, which is an approximation of a 50Ω termination; and then into the ON state, whereby an active noise power with a defined value is created. The ENR factor ("Excessive Noise Ratio") represents the relation between the noise power density in the ON and OFF states. It is measured in [dB] and may be frequencydependent. In practice, noise sources are implemented with a Read diode as a generator in combination with a downstream attenuator.
Since the complex reflection factor of the noise generator is different for each of the two switch states, it is necessary to determine the noise reflection factor of the noise source for both switch states first in order to obtain exact measurements of the noise figure. With these measurement values, it is then possible to correct the noise figure. The goal of the correction is to obtain a valid noise figure for a circuit with a source load of precisely 50Ω. If an ideal 50Ω source load is assumed, the noise figure for an ambient temperature of 20°C can be calculated in [dB] as:
where the noise power density ratio measured at the amplifier output for the ON and OFF states, Y_{L}, is defined as:
An alternative method for measuring noise figures is the socalled "Cold Source" method. If this method is used, it is also necessary to determine the exact value of the test amplifier gain, since this will now be included into the calculation. Therefore, it always makes sense to measure the gain and noise figure of the amplifier under test at the same time, without having to change the setup of the measuring station.
Let us now have a look at the case of a noise source with variable source impedance, which can be adjusted by means of a tuner. The ohmic losses of the tuner and feed lines and the reflection factor of the noise source are not taken into account. The noise figure of the amplifier can now be analytically represented as a function of the source impedance (Z_{0 }is the characteristic wave impedance of 50Ω; Γs is the complex reflection factor of the source, Γopt is the complex reflection factor of the source for optimum noise matching, when the minimum noise figure of Fmin is achieved; Rn defines the steepness of the noise figure increments):
This means that four noise parameters are required in order to describe the noise behavior of an amplifier: Fmin, Rn, the real component {Γopt} and the imaginary component {Γopt}. This formula is extremely useful for practical purposes, since it describes the increase of the noise figure in case of a mismatch. A mismatch may occur due to unsuitable matching circuits at the amplifier input, or due to tolerances of the passive components of the matching network/the active semiconductor circuit. Figure 2 shows an example noise figure distribution calculated with the formula described above. If this graphic is viewed from above (and thereby reduced to two dimensions), the Nyquist plots for constant noise figures are visible as circles around the point of optimum noise matching, Γopt.
If the ohmic losses and the reflection factor of the noise source are taken into account, the process to evaluate the noise figure becomes very elaborate. In this case, it is not sufficient to use just the Y factor method or the cold source method. Maury Microwave has implemented a numerical method in the ATSv5 Software to apply these corrections. This requires two calibrations with the network analyzer before performing the noise measurement: one 2portcalibration for both reference planes of the amplifier under test and one 1portcalibration for the reference plane of the noise source. The noise parameters are then obtained by using a fitting procedure on the measurement data.
The above photograph shows a noise measuring station which uses a network analyzer with optional internal HF switches and a vectoral noise receiver. The internal HF switches allow switching between measuring the S parameters and the noise figure. With a further option of the network analyzer, this measuring station can support additional measurements of the 2tone intermodulation performance of the amplifier under test.
We recommend the components listed in the following table for automatic noise measuring stations which are used to obtain noise parameters:
Noise source 
For amplifiers with a gain of 1020dB, we recommend noise sources with an ENR of around 15dB. 

Tuners 
Motorized mechanical tuners are ideal, e. g. by Maury Microwave. 

Amplifier under test  The amplifier component to be tested will be contacted via a test socket or via the wafer prober. Possible measurement falsifications which are due to electromagnetic interference can be prohibited by using special shielded housings with EMCproof cable ducts for power supply and HF cabling.  
Noise receivers  For this purpose, highly sensitive spectrum analyzers may be used. A very efficient alternative are the PNAX network analyzers by Keysight, equipped with option 029 (vectoral noise measurement) and internal HF switches for switching between measuring the gain and the noise power. If a PNAX is used, it is possible to integrate the 2tone intermodulation measurement with additional options.  
Measurement cables 
As measurement cables, we recommend phasestable, lowloss measurement cables, e.g. W.L. GORE or Maury Microwave. It is a basic requirement to place the noise source as closely as possible to the DUT or the tuner. However, this is not always possible when a waver prober is used, and lighter HF cables may have to be used for contacting the DUTs. 

HF switches at the input of the amplifier under test  The best option are lowloss, reliable HF switches which can be controlled by software.  
Preamplifier at the output of the amplifier under test 
In many cases, the gain of the amplifier under test or the input sensitivity of the spectrum analyzer will not be sufficient to measure the noise level correctly. In these cases, an additional lownoise preamplifier is required. 

Software 
The proven ATSv5 Test SW by Maury Microwave is recommended as a controller and for obtaining the measurement data, including the following modules: MT993B for basic noise measurements:
MT993B01 for ultrafast noise parameter measurements
The following additional modules can assist in making measurement tasks extremely efficient: MT993E for external control of the ATS SW
MT993F System Control SW
MT993G for displaying DC currentvoltage characteristic curves
MT993J for characterizing the test fixture

The following examples show the measurement results for two different amplifiers under test:
 SGA4363Z: Repeater by QORVO. Well matched to 50Ω at the input and at the output.
 BFP193: Discrete HF transistor by Infineon Technologies AG. Not matched at the input; the output has an impedance of approx. 50Ω.
Both amplifiers have been mounted in a shielded metal housing and are powered by battery in order to minimize any influence by external interferences (for the measurement setup, see Figure 3). The ATSv5 SW by Maury Microwave with modules MT993B and MT993B01 was used to capture the measurement values and to determine the noise parameters. The picture below shows how the individual components in the configuration menus of the ATSv5 SW were adapted to the available measurement instruments:
The results for the noise parameters of the SGA4363Z are shown in Figure 5. The noise parameters were determined in a frequency range between 0.83.0GHz, with an increment of 100MHz. The righthand side of the picture shows Rn, Fmin, Associated Gain and Gmax. The maximum gain, Gmax, was calculated; it would be reached in case of an optimum match of the amplifier output to 50Ω. The lefthand side of the picture shows the source impedance Γs in the Smith Chart for each individual frequency point. If a cursor is placed on a frequency point, it is possible to view the circles for a constant noise figure. In our example, a frequency of 1.0GHz was selected.
The result shows that the SGA4363Z makes an excellent repeater. The minimum noise figure that was reached is very close to a source impedance of 50Ω; the increase of the noise figure in case of a mismatch is relatively low. The noise figure stays at an acceptable level within the inner area of the Smith Chart for a large range of source impedance values. The specified noise figures for the SGA4363Z are 2.7dB at 850MHz/3.1dB at 1950MHz. The measured noise figures are a bit higher; this is due to the losses caused by the circuit board and the integration of the component into the shielded housing (e.g. 3.41dB at 800MHz and 3.94dB at 1900MHz).
The results are completely different for the BFP193. The minimum noise figure of 3.05dB is reached at a source impedance of 8.0 + j · 13.4Ω (Γ_{L}= −0.64 — j · 0.38Ω). However, the noise figure rises to more than 6dB at a source impedance of 50Ω. If this amplifier will be used in a 50Ω environment, the input impedance needs to be adapted accordingly. In most cases, this is achieved by circuits with discrete coils and capacitors. However, this may cause problems, since in many cases a good adaptation can only be achieved for a frequency bandwidth of 1020% of the center frequency. In this case, we were also unable to reach the theoretically achievable noise figure of the BFP193 (1dB at 900MHz/1.6dB at 1800MHz), since the ohmic losses caused by our test circuit board, the DC power source and the shielded housing were too high.
The ATSv5 SW also supports displaying circles for a constant associated gain (see Figure 7). This is very helpful for a quick visual analysis of the amplifier behavior. It is also possible to evaluate the measurement data in retrospect. This allows recording measurement data sets automatically overnight and evaluating them during normal operating hours. It is thereby possible to make optimum use of the expensive measurement equipment.